Some features of beckman.com will be unavailable from 8:00 p.m. PDT Saturday, December 7 until 12:00 a.m. PDT, Sunday, December 8 for planned maintenance. During this time, you will not be able to log in to use account features such as online shopping, shopping lists and order tracking.
Please ensure your orders are processed prior to this maintenance period or contact Client Services for additional assistance.
What is the difference between instrument verification and instrument calibration?
For MET ONE and HIAC particle counting equipment, the CoreCAL calibration system is required to make adjustments that may be needed to maintain resolution and counting accuracy to industry standards. Performing an instrument verification following calibration adjustment will confirm if your instrument is performing within manufacturer’s tolerance CoreCAL also gives an “As Received” data as part of a full calibration), to inform you as to what extent the instrument has drifted since its last calibration Audit quality CoreCAL calibrations are available through a Beckman Coulter Service Network.
Oops. The page you wanted could not be found, so we brought you to something similar. If you still can't find what you want, try searching our site instead.